Reciprocal Levelling | Curvature and Refraction Correction
Reciprocal levelling helps in compensating for the error due to curvature and refraction & also line of collimation errors in surveying.
In reciprocal levelling, the level is set up on both sides of the levels. Two sets of staff reading is taken. This helps in compensating for the error due to curvature and refraction & also line of collimation errors in surveying.
It is assumed that the atmospheric refraction remains at the same level when the instrument is shifted (Advisable to shift the instrument quickly!)
Curvature and refraction error
Curvature correction : Horizontal line is not a level line, because of which the staff reading is more than expected. Negative correction required.
Refraction correction : Rays of light passing through the atmosphere of different density bends down. Positive correction to be applied.
Reciprocal Levelling Steps and Formula
APSEd is an E-learning platform for Civil Engineering by IIT-B Graduates. Get access to all in one place!
Video Lectures & Study Material
Practice Questions & Mock Tests
Exclusive Doubt Support
600+ learners are studying with us. Join them and learn with us for an optimized learning experience.
Hi! I am Aditya, co-founded APSEd with Palaniappan.
Our brain can accumulate and memorize things to a certain extent. Once it is overloaded, some of the facts and formulas you remembered fade away over time.
All this becomes easier once we understand the concept and know how to apply them. This is what we strive to achieve through our courses.
Our Courses + Your Effort = Success in exams!
Let us say A and B are two points as shown in figure below.
Instrument is setup near A and readings on the staff are noted (x1 and y1)
Instrument is shifted near B and readings on staff are again noted (x2 and y2)
Let the curvature and refraction error be e, the calculation is as shown below